Introduction of MANI DIA-BURS
Wide variety of diamond burs with a code indicating the shape of the bur.
Color code indication is provided at the end of working part ( ), according to the size ofdiamond grit conforming to the specification.
MANI coding system reflects shapes of burs. The contour and sizes could be imagined by MANI nomenclature since ISO numbers arealso indicated. ●TF/Taper Flat End ●TR/Taper Round End ●TC/Taper Conical End ●FO/Flame Ogival End ●SF/Straight Flat End ●SO/Straight Ogival End ●SR/Straight Round End ●RS/Rounded Shoulder ●EX/Special (Extra) Shape ●BR/Ball Round Type ●BC/Ball Collar Type ●DI/Double Inverted Cone ●SI/Single Inverted Cone ●WR/Wheel Round Edge ●CR/CR Inlay Preparation Dia-Burs ●CD/Children's Dentistry Dia-Burs ●S/Short Shank ●SS/Super Short Shank ●C/Coarse ●F/Fine ●EF/Extra Fine
High Quality Control
Laser measurement cutting test and strength test for each lot are performed. All products comply with our own standards whichare far more strict than ISO standards.
Before using MANI DIA-BURS, please see the instruction as indicated below.
●For dental use only.●Non-sterile.●Insert the shank into the handpiece according to the instruction of the handpiece manufacturer.Insertion should be the full length.●Beforeuse, run the instruments to make sure there is no descentering. If the instruments become bent, be sure to remove them from anyfurther use.●Please always use spray. APlentiful water supply is necessary to maintain long-lived performance and also to prevent loading and pulpitis.●Handle with care not to put excessive load. Pressure should be20 to 50p for all instruments with diameter up to 023.●Use below max. allowable r.p.m. of each instrument, see chart above, to avoid a possibility of being injured due tobreakage.●Avoid levering, tilting, twisting and jamming motion. Do not work with a pulling motion.●Wear a safety glasses and/or facemask to protect from being injured.●Practicable for using auto-clave. Do not use high pressure steam sterilizer which heats more than 200°Cincluding drying process.●After using, follow appropriate treatment as themedical waste.
- Yinzhong Xu